Like any high-tech device, hearing aids may occasionally become fussy. Not to worry – most issues can be solved easily and at home. Below are some of the most common calls we get at our Support line, and the simple troubleshooting tips you can do to fix them from the safety of your home.
If you experience one of the following problems:
- Loss of sound
- Weak or intermittent amplification
- Internal noises or distorted/garbled sound
Then ask yourself these four questions to troubleshoot.
1) Is my hearing aid on?
If you aren’t getting any sound from your hearing aid, ensure the hearing aid battery door is completely closed and the hearing aid is turned on. If you have a volume control, be sure the volume is not set on mute or on the minimum setting.
2) Is there a problem with the battery?
The battery needs to sit in the hearing aid in a specific way for it to work (refer to your instruction manual for information on proper placement). Like other batteries, hearing aid batteries are polarized and have both “+” and “-” markings. The flat side of the hearing aid battery is the positive pole. Check for proper placement of the battery in your hearing aid. If the battery is upside down, your hearing aid won’t be able to draw power.
Next, make sure you aren’t working with a dead or weak battery. Remove a fresh battery from the package and remove the sticker/tab from the battery. We recommend you wait 3-5 minutes after removing the sticker before using the battery. This time allows the battery to completely charge before being ready for use. Insert the new battery into the hearing aid. If you have a battery tester, you can use it to check the power level of the hearing aid battery as well.
3) Is the hearing aid clogged?
Earwax, moisture, oils and other foreign materials can build up in the microphone(s), sound port, and tubing of the hearing aid negatively affecting its function.
- Wipe hearing aid with a clean, dry cloth or tissue.
- If your hearing professional provided you with a cleaning tool or brush, use it to clean debris from the hearing aid’s components.
- Clean or replace wax guard/protection.
4) Am I wearing the most appropriate program?
If your hearing aid has multiple programs, verify that you are using the most appropriate program for your environment. Some programs that are intended for the telephone or specific environments may sound unusual if you are using them in other situations.
If all of these efforts fail, then contact us at Purchase Ear Technology. Purchase Ear Technology in Paducah, KY is your one-stop for everything hearing aid related. From hearing tests, new hearing aids, service and repair, we are here to help. Our office is located at 2008 Broadway, Paducah, KY. You may reach us by phone at (270) 558-3996 or CLICK HERE to contact us. At Purchase Ear Technology, you are more than a customer. You are FAMILY!